Iterative decomposition of water and fat with echo asymmetry and least-squares estimation contrast-enhanced T1-weighted and T2-weighted imaging effectively reduces the degree of tissue-obscuring MR imaging artifacts produced by spinal fixation hardware and improves image quality compared with conventional sequences.
To prospectively compare the effectiveness of iterative decomposition of water and fat with echo asymmetry and least-squares estimation (IDEAL) T2-weighted and contrast material–enhanced T1-weighted magnetic resonance (MR) imaging with that of a conventional MR imaging protocol in minimizing metallic artifacts in phantoms and in patients with metallic hardware after spinal surgery.
Materials and Methods
Institutional review board approval and informed consent were obtained for this study. Coronal T1- and T2-weighted MR images of six titanium alloy pedicle screws in an oil bath containing tubes filled with diluted gadolinium contrast medium were obtained with frequency-selective fat saturation (FSFS) and IDEAL. Axial T2-weighted and contrast-enhanced T1-weighted MR imaging of the spine was performed with FSFS and IDEAL at 22 lumbar levels in 19 patients. Two musculoskeletal radiologists qualitatively analyzed the images in terms of the visualization of paravertebral muscle and the spinal canal region, uniformity of fat saturation, and noise. The paired images were rated by using a five-point scale. For the quantitative study with phantoms, the short- and long-axis lengths of metallic artifacts were determined on signal intensity profiles.
In the phantom study, metallic artifact size was markedly decreased in the IDEAL T2-weighted and contrast-enhanced T1-weighted images (P < .001). In the clinical study, IDEAL T2-weighted and contrast-enhanced T1-weighted images enabled significantly improved visualization of the dural sac (P < .001), spinal muscles (P < .05), uniformity of fat saturation (P < .001), and noise (P < .05).
IDEAL T2-weighted and contrast-enhanced T1-weighted MR imaging examinations effectively reduce the degree of tissue-obscuring artifacts produced by spinal fixation hardware and improve image quality compared with FSFS T2-weighted and contrast-enhanced T1-weighted MR imaging.
© RSNA, 2011
Supplemental material: http://radiology.rsna.org/lookup/suppl/doi:10.1148/radiol.11101856/-/DC1
- 1 . Lumbar spine fusion and stabilization: hardware, techniques, and imaging appearances. RadioGraphics 2007;27(6):1737–1749. Link, Google Scholar
- 2 . Optimizing imaging parameters for MR evaluation of the spine with titanium pedicle screws. AJR Am J Roentgenol 1996;166(5):1213–1218. Crossref, Medline, Google Scholar
- 3 . IDEAL imaging of the musculoskeletal system: robust water fat separation for uniform fat suppression, marrow evaluation, and cartilage imaging. AJR Am J Roentgenol 2007;189(5):W284–W291. Crossref, Medline, Google Scholar
- 4 . Metallic artifacts on MR images of the postoperative spine: reduction with fast spin-echo techniques. Radiology 1994;190(2):565–569. Link, Google Scholar
- 5 . MRI of spinal hardware: comparison of conventional T1-weighted sequence with a new metal artifact reduction sequence. Skeletal Radiol 2001;30(4):213–218. Crossref, Medline, Google Scholar
- 6 . Iterative decomposition of water and fat with echo asymmetry and least-squares estimation (IDEAL): application with fast spin-echo imaging. Magn Reson Med 2005;54(3):636–644. Crossref, Medline, Google Scholar
- 7 . Multicoil Dixon chemical species separation with an iterative least-squares estimation method. Magn Reson Med 2004;51(1):35–45. Crossref, Medline, Google Scholar
- 8 . High-resolution 3D cartilage imaging with IDEAL SPGR at 3 T. AJR Am J Roentgenol 2007;189(6):1510–1515. Crossref, Medline, Google Scholar
- 9 . Measurement of signal-to-noise ratios in MR images: influence of multichannel coils, parallel imaging, and reconstruction filters. J Magn Reson Imaging 2007;26(2):375–385. Crossref, Medline, Google Scholar
- 10 . MR evaluation of stent patency: in vitro test of 22 metallic stents and the possibility of determining their patency by MR angiography. Invest Radiol 2000;35(12):739–746. Crossref, Medline, Google Scholar
- 11 . The measurement of observer agreement for categorical data. Biometrics 1977;33(1):159–174. Crossref, Medline, Google Scholar
- 12 . Overcoming artifacts from metallic orthopedic implants at high-field-strength MR imaging and multi-detector CT. RadioGraphics 2007;27(3):791–803. Link, Google Scholar
- 13 . Artefacts in magnetic resonance imaging caused by dental material. MAGMA 2005;18(2):103–111. Crossref, Medline, Google Scholar
- 14 . Metallic artefacts in MR imaging: effects of main field orientation and strength. Clin Radiol 2003;58(4):322–328. Crossref, Medline, Google Scholar
- 15 . Improved MR imaging for patients with metallic implants. Magn Reson Imaging 2000;18(3):287–295. Crossref, Medline, Google Scholar
- 16 . MR imaging of the liver using short TI inversion recovery sequences. J Comput Assist Tomogr 1985;9(6):1084–1089. Crossref, Medline, Google Scholar
- 17 . Quantitative assessment of an MR technique for reducing metal artifact: application to spin-echo imaging in a phantom. Skeletal Radiol 2001;30(7):398–401. Crossref, Medline, Google Scholar
- 18 . Technical considerations: CT and MR imaging in the postoperative orthopedic patient. Semin Musculoskelet Radiol 2002;6(1):5–17. Crossref, Medline, Google Scholar
- 19 . Postoperative lumbar spine: contrast-enhanced chemical shift MR imaging. Radiology 1991;179(2):557–562. Link, Google Scholar
- 20 . T1- and T2-weighted fast spin-echo imaging of the brachial plexus and cervical spine with IDEAL water-fat separation. J Magn Reson Imaging 2006;24(4):825–832. Crossref, Medline, Google Scholar
Article HistoryReceived September 25, 2010; revision requested October 27; revision received November 23; accepted December 3; final version accepted December 15.
Published online: June 2011
Published in print: June 2011